UPDATE 1-U.S. FDA warns Zimmer over manufacturing of hip devices
Oct 1 (Reuters) - Zimmer Holdings Inc said it received a warning letter from the U.S. Food and Drug Administration citing problems with manufacturing and testing processes for artificial hip devices made at the company's Ponce, Puerto Rico, plant.
The warning letter, dated Sept. 19, does not restrict production or shipment of the hip devices, called Trilogy Acetabular Systems, or require withdrawal of any products from the market, Zimmer said in a regulatory filing.
The FDA said Zimmer failed to implement a testing mechanism to demonstrate the products met certain design specifications and had not validated the effects of manufacturing operations on products that incorporate metallic spikes.
The warning follows a June inspection of the company's Ponce plant. Zimmer said it temporarily suspended release of the affected products and has since completed the validation process for Trilogy products that do not incorporate metallic spikes.
The company expects to complete validation of Trilogy spiked products within the next few weeks.
- Islamic State executes soldiers, takes hostages at Syria base: social media
- WHO shuts Sierra Leone lab after worker infected with Ebola
- Gaza truce holding but Israel's Netanyahu under fire at home |
- Talks optimism fades as Ukraine says Russia makes new incursion |
- 'Men in green' raise suspicions of east Ukrainian villagers